Real time monitoring of lung function is of particular interest to the mechanically ventilated patients during critical care. In most cases, the patients remain immobilized in supine position for long periods of time. As a result of this recumbent position, a marked amount of fluid is accumulated in the posterior parts of the lungs due to gravity. Unless diagnosed early and treated successfully, the fluid accumulation leads to impaired gas exchange and may cause a respiratory failure. One common way to diagnose the fluid accumulation in the lungs is to have an x-ray examination of the patient, still, it requires the transport of the patient and does not allow for real time monitoring. Despite its low spatial resolution, electrical impedance tomography (EIT), a noninvasive and inexpensive device, is regarded as a promising approach to continuously monitor the ventilation distribution . The technique is based on the fact that, during ventilation, the lungs are periodically filled up with air which results in an increased impedance at the regions where the lung tissue is present. Therefore, local impedance variations indicate that the lung tissues at these regions are normally functioning. That way, EIT images has the potential to guide clinicians to optimize lung recruitment and maintain a reliable medical prognosis. However, notwithstanding the considerable improvements in the last decade, using electrodes in direct contact with the skin yet induces serious problems with regard to the poor skin-electrode contact  and the electrode position uncertainties . Besides, depth sensitivity gets worse if there exists a poorly conducting layer, i.e., bone tissue, on the surface . Additional side-effects of using electrodes may be irritated skin or even an allergic reaction, particularly for the infants and patients with sensitive skin.
Recent studies have shown that magnetic induction measurements with single coils provide signals which are correlated with the lung dynamics . However, for regional monitoring, in principle, it is essential to collect a set of measurements from different sites to be used in image reconstruction. Therefore, this idea is extended here by using coil arrays for data acquisition in order to image the regional thoracic conductivity changes. By doing so, the contact problems faced in EIT can readily be eliminated and the patient comfort can be improved. Ona major drawback of using coils is the image degradation caused by the uncertainties of the skin positioning with respect to the coils . The proposed sensor configuration which is comprised by 25 coils in a 5 by 5 planar arrangement is illustrated in figure 1. The sensor system is assummed to be mounted onto the patient bed and will be used to expose the body and record the measurements. With this configuration, the measurement cables can easily be installed through holes drilled in the bed.
Although, the physical nature of the problem is equivalent to that of magnetic induction tomography (MIT) [7, 8], which is another impedance imaging modality alike EIT, the sensor configuration differs significantly from MIT. Considering also that the principal focus is monitoring of the regional lung dynamics, the proposed technique will be referred to as magnetic induction pneumography (MIP). In MIP, the data are acquired by successively exciting each coil in order to induce an eddy-current density within the dorsal tissues and measuring the corresponding response magnetic field strength by the remaining coils. The recorded set of data is then used to reconstruct the internal conductivity distribution by means of algorithms that minimize the residual norm of the difference between the estimated and measured data . Regularization methods are typically applied to stabilize the image reconstruction process .
To simulate the measurements, an eddy-current finite element solver was used to compute the electromagnetic fields. The sensitivity maps were obtained by computing the adjoint fields according to Geselowitz’s reciprocity theorem . For the assessment of the technique, the sensitivity maps for each transmitter-receiver coil pair were studied. The spatial resolution of the configuration was investigated by examining the point spread functions (PSF) at different locations and depths. To simulate a realistic scenario, a chest model was generated by segmenting the tissue boundaries from NMR images. The electrical properties were obtained from the literature . Truncated singular value decomposition (SVD) was performed in order to reconstruct the images. The reconstructions of the ventilation distribution and the development of an edematous lung injury were presented. The imaging artifacts caused by either the incorrect positioning of the patient or the expansion of the chest wall due to breathing were studied by simulations.
The forward problem is defined as, given the electrical properties of the body and the geometry of the coil system, solving for the induced voltages in the receiver coils. This requires the solution of Maxwell’s electromagnetic equations which was reported in several earlier studies (see, for example ). A brief review is given here to provide a background and set the stage for the inverse problem.
Assume an AC driven coil located in the proximity of a volume conductor which is represented by Ω. The forward problem is to compute the induced voltages in the remaining pick-up coils due to the eddy-currents in Ω. According to Geselowitz’s reciprocity theorem, the induced voltage can be expressed as follows,
In MIP and all related impedance imaging techniques, applying a differential measurement strategy is favorable because the systematic errors and drifts are better suppressed in this way. However, if the measurements are taken at different operating frequencies, unlike EIT, a scaling is necessary  as given below,
Presently linearization approaches are commonly used to deal with the nonlinearity of the forward map. By linearizing (1), the change of voltage data, ᐃ
When a piecewise constant conductivity distribution, i.e., σ = [σ1,σ2, · · · ,σ
The entries of
The image reconstruction problem is to restore the conductivity perturbation,
Let ρ1 be the largest singular value and ρ
Assuming that the measurement data are uncorrelated and the data variances are normalized to unity, the image resolution can be expressed in terms of the SVD components as follows ,
Thus, the resolution matrix is simply
A 5× 5 planar coil arrangement used in simulations is depicted in figure 1. The solenoid coils were approximated as infinitely thin circular wires. They are assumed to act both as transmitters and receivers and have a radius of 2.5 cm. The sensors cover a square region of 625 cm2 in total. For the simulations, the approximate distance from the coil center to the body surface was 1 cm. Figure 3 illustrates the coil geometry used in simulations with respect to the 3-D chest model.
The tissue boundaries of the chest model was segmented from the NMR images to construct a three dimensional volume. Typically, reconstruction of small details is infeasible with diffused imaging modalities; therefore, the model was constructed roughly by considering only the extended tissue groups such as the lungs, heart and the surrounding muscle tissue. Two different tetrahedral meshes with different sizes of approximately 100000 and 60000 elements were generated to be used independently for voltage simulation and image reconstruction, respectively. The electrical conductivities of the tissues were taken as 0.27, 0.11, 0.21 and 0.36 Sm−1 for deflated lung, inflated lung, heart and muscle tissues, respectively, corresponding to 100 kHz operating frequency .
The fluid accumulation that causes edematous lung injury takes usually place when the patients spend long time immobilized in supine position, and thus, gravitation forces a marked amount of fluid accumulate in the posterior parts of the lungs. The collapsed region is commonly a mixture of edematous fluid together with the deflated lung tissue causing the conductivity of that region to be higher than the healthy tissue. A conductivity value of 0.6 Sm−1 was assigned to the collapsed region according to . The formation of edema in the posterior regions of the lungs is illustrated as red regions in figure 3.
The sensitivity matrix of the design was calculated assuming that the coil plane is located 1 cm above a half-space that has a uniform conductivity distribution. The sensitivity distributions for different coils in the sensor matrix are presented in figure 4. Both the sensitivity distribution near the surface and on the transverse plane are shown. For all setups, the region between the coils provides positive voltages, i.e., positive sensitive regions. The black lines represent the height and width of the regions where the sensitivity drops to 95% of the maximum sensitivity. The depth sensitivity increases with the increasing relative distance between the coils. However, further increase of the distance than 15 cm did not show any improvement in depth sensitivity.
The resolution matrix was computed using the same half-space model by assuming that the coil plane is located 1 cm above the half-space. To determine the truncation level necessary for a stable image reconstruction, the logarithmic ratio between the first and the remaining singular values as compared to the SNR levels was examined. The optimum truncation level for the designs corresponding to the SNR levels of 40 dB, 60 dB and 80 dB were found to be 145, 216 and 275, respectively. The PSFs of several locations for the SNR level of 60 dB are presented in figure 5. Near the surface regions, the PSFs are sharp and narrow, however, with increasing depth, they become wider and show more diffuse characteristics. Also the maximum amplitude of the PSFs approximately drops exponentially with depth. Besides, the peak of the PSFs shift towards the surface regions (more in deeper regions and less in surface-near regions) and thus, the localization errors become more prominent with increasing depth.
Assume that the patient is in a supine position and the measurements are taken at different times,
Is some cases, especially for the patients that spend long time in a supine posion, a markedly increased gravitational fluid accumulation hinders normal functioning of the lungs in the posterior regions. To simulate this, it was assumed that the fluid accumulation has reached a state in which 2.5 cm, 5 cm and 7.5 cm of the lower lungs have completely collapsed (refer to figure 6 for the illustration of levels). The differential measurements at
It is also possible to image the edematous lung injury by recording the initial measurements from the patient with healthy lungs at
Figure 10, shows the normalized edema reconstructions along a line passing through the peak value of the reconstructed edema from ventral to dorsal, normal to the coil array plane, (see the red line in figure 9). The x-axis represents the ventral to dorsal orientation. The corresponding full-width-at-half-maximum (FWHM) values of the curves are illustrated as boxcar functions. It was found that the FWHMs provide marginal information on the level of the collapse, however, it is clearly visible that the amplitude drops drastically with the increasing volume of fluid accumulation. Thus, the variations in amplitude can be used to infer the development of fluid accumulation rather than the distributions.
It is inevitable that the patients change the posture or position during ventilation which causes alterations in the measurement signal. It was previously reported that these alterations are noticed as serious artifacts in the images . To this end, we performed a simulation test to investigate the stability of image reconstructions to several body distorsions. The differential measurement were simulated by assuming that the body posture is modified at
The linearity assumption for the solution of the inverse problem may lead to undesired conductivity reconstructions when the change of the conductivity distribution is too large. This seems an important issue, particularly for lung applications, where the lung tissue is distributed extensively and the perturbations hardly fulfill the linearity assumption. Applying more iterations (i.e., Gauss-Newton iterations) may not always fix the problem, since the iterative algorithm usually converges to a local solution which may not be the desired one. A good strategy to overcome this problem is to select better conductivity distributions to be used for the computation of the sensitivity matrix. So far, uniform distributions have been used to attain the sensitivities, however by doing so, the imaging quality was found to be insufficient to clearly distinguish the right lung from the left lung . In this study, the sensitivity matrix was calculated by using an inhomogenous model, the tissue boundaries of which were segmented from NMR scans of the patient. Since many patients undergoing monitoring usually have these scans beforehand, this approach can easily be adopted. Such prior knowledge, as emphasized in this paper, increases the image quality, though, still severe artifacts can be noted especially near the boundary of the images.
The image reconstruction problem is severly ill-posed as can be noticed from the rapid decay rate of the singular values. Hence, regularization methods must be employed to overcome the ill-posedness and stabilize the image reconstruction. In this work, the truncation of the singular values was used for this purpose, however, there are several other approaches which have the potential to regularize the problem more efficiently. For instance, since the data acquisition is rapid and the successive data frames are strongly correlated, an appropriate strategy would be implementing a “Kalman filter” to take advantage of the temporal correlation in data frames within a reasonable time span . On the other hand, another way to improve regularization would be to use additional information based on the variability of the physiological parameters. It is known that the conductivity of the tissues varies with gender, age and condition of the patients, which renders it essential to determine the corresponding statistics to the model and to include this information as a
It was previously reported for a complete magnetic induction tomography system that the artifacts due to breathing usually spoil the images . However, for the patient in a supine position, the chest motion due to respiration is assumed to take place only on the ventral surface and the dorsal movement artifacts are expected to be minimal for this position. This argument has been explored here and the artifacts due to breathing is found to be not as severe as in the complete tomography system. However, rotational and transversal displacements of the patient during data acquisition still cause significant artifacts. Nevertheless, considering the patients that are anestesized and stay immobilized in the ICUs, these rotational and displacement artifacts are not primarily influential. Thus, the images obtained with this planar system appears to be less sensitive to the motions than the complete tomography system.
In this work, the measurements were assumed to be insensitive to the cardiac cycles. The heart is placed in the ventral portion of the chest and, knowing that the sensitivity decays rapidly with depth, it is reasonable to neglect the corresponding effects. This can also be verified by analyzing the PSFs of the system. A conductivity perturbation at the location of heart shows a very distributed and smooth PSF with a very low amplitude and does not have a significant effect on the reconstructions of the dorsal regions. Nevertheless, it may be useful to apply a band-pass filter to supress signals correlated with the heart rate.
The feasibility of lung imaging using a planar coil matrix was explored. A realistic thorax model was constructed using NMR images and used for the analysis. This preliminary work points out the potential of the method to be used for real time monitoring of ventilation and fluidic changes. The sensor system can easily be mounted onto the patient bed. With this planar system, the sensitivity of the measurements drops rapidly with depth below the skin, therefore, the inhomogeneities in deeper regions can hardly be monitored. However, unless supressed via a filtering strategy, severe imaging artifacts can be induced due to the patient’s movement during data acquisition. For immobilized patients, the artifacts due to breathing is found to be not as severe as the complete tomography system. The future work may include finding optimal coil matrices so that the spatial resolution of the images is increased for particular locations which may provide better diagnostic information.